Auto-suggest helps you quickly narrow down your search results by suggesting possible matches as you type. The VI instrument driver works. Error 6 I get it only when pressing “Take Measurement” button. The data I want to represent is IV curves. At the same time I get an error on the monitor of D00 After this I can see the cursor on the bottom of the I V graph on the monitor of the sliding from left to right when settinp a Sweep of Voltage of SMU1 from 0 to 1 V, pts.
|Date Added:||1 October 2011|
|File Size:||65.39 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
Message 6 of The data I want to represent is IV curves.
You must also provide an include path for hpb. You can right click on the error cluster which shows error 6 and select “Explain Error”. Do you have any experience with this? All of the directory paths which follow assume Windows This is on the B, right? It includes a comprehensive on-line help file which complements the instrument manual.
b free download, or read b online
I have a very old VI with a very old driver. I looked on the manual for this error and I cound not find it.
Here should be what you are looking for. I will write Keysight, even though this product I believe to be discontinued, maybe they can offer me some support. First, there is no soft front labviea, as the instrument can be controlled from its hardware front panel.
Solved: LabView VI for HP : diode measurement IV – Discussion Forums – National Instruments
At the same time I get an error on the monitor of D00 Did you try other commands which take a longer time to execute? There are lqbview advantages to using the dll form of the driver. Message 1 of 6. Message 8 of I am new to the forum and also new to LabView.
Next step I have tried before preparing the probestation was to set for one of the SMUs SMU1 a voltage value and measure the set value using a multimeter. Message 3 of 6.
Agilent 4156 Analyzer
Message 10 of I managedto get communication between the equipment HP and LabView. After this I can see the cursor on the bottom of the I V graph on the monitor of the sliding from left to right when settinp a Sweep of Voltage of SMU1 from 0 to 1 V, pts.
It includes a Visual Basic include file. I would like to know if there is any VI available for 2 4156g devices in this case on wafer diodes so that I can control directly the equipment from LabView and be able to read out the measured data directly from the HP via LabView?
The VI instrument driver works. Sampling on HP If you do not have internet access, the drivers can be obtained by contacting your country support organization. This is providing the source voltages and also measuring through the SMUs perfectly.
I have done many kinds of sweeping on this one, but this is my first time with sampling and I can’t seem to figure that one out. Norbert Norbert CEO: The rest of the channels are not assigned, as now I only try one port measurement, to see if I can take out data from the HP You can specify either a total sampling time or a specific number of points.
However, for maximum portability, labvlew instrument address should use uppercase characters only.
Error 6 I get it only when pressing “Take Measurement” button.